Read online or download PDF • Page 44 / 67 • KROHNE IFC Converter EN User Manual • KROHNE Equipment. Micro Precision provides expert calibration services for KROHNE instruments including KROHNE IFC D -. Select from options for the IFC D which. KROHNE 08/ Subject to change without notice. Applicable to. Software-Versions. ○ IFC _. / D. Display version. No. and.
Page 15 Function 3. Tell us about it. V ariable area flowmeters.
KROHNE IFC Converter EN User Manual | Page 44 / 67
Page 8 Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical e of outputs 2. Page 14 Function 1.
Do not cross or loop cables inside the terminal compartment. Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical connection of outputs 2.
Switches, counters, displa ys and recorders. The selected creepage distances and clearances have been sized in conformity with VDE kroyne IEC for contamination category 2. Or point us to the URL where the manual is located.
Connect shields via stranded drain wires.
Underwater and underground installation possible. P Pulse output overranged. Supply circuits and output circuits are designed to meet the standards of overvoltage classes III and II, respectively. Power the unit, and the flowmeter will immediately start process flow measurement. Krobne concise instructions do not include the following: The flowmeter housings protecting the electronic equipment from dust and moisture must always be kept closed.
If installed and operated in accordance with these operating instructions, it will rarely present any problems.
Page 2 Contents 1 1. Tell us what’s missing. Passive mode External power source required for operation of the outputs. The operator is, howeverobligated to take note of these sections in the. Signal cable to be permanently laid.
Vo r tex flowmeters. Page 11 Standard factory settings Function 1.
Your manual failed to upload Page 10 3 Start-up 3. Setting ranges The ranges are dependent on the meter size DN and the flow velocity v: Page 3 1 Electrical connection: Lev el measuring instruments. Don’t see a manual you are looking for?
Use separate cable entries see below for power supply, field current cables, signal lines, outputs and inputs. All operating data have been factory set in accordance with your specifications.