80T-IR – read user manual online or download in PDF format. Pages in total: Find great deals for Fluke 80t-ir Temperature Probe 80TIR. Shop with confidence on eBay!. Find great deals for Fluke Infrared Temperature Probe M# 80t-ir. Shop with confidence on eBay!.
Save to parts list Save to parts list. May substitute less accurate blackbody if using in conjunction with transfer standard of equivalent accuracy, i. Reactivate by switching the UUT off and on again. Search the history of over billion web pages on the Internet.
Save to an existing parts list Save to a new parts list. The foregoing information relates to product sold on, or after, the date shown below. Power up the UUT and allow it to stabilize for five minutes. Repeat steps 1 through 5 for the remaining verification points called out in Table 2. Place the UUT aperature three inches from the center and perpendicular to both axes of the Blackbody plate. Sending feedback, please wait Thank you for your feedback.
Refer to Figure 1 for equipment configuration. The Manufacturers disclaim all warranties including implied warranties of merchantability or fitness for a particular purpose and are not liable for any damages arising from your use of or your inability to use the Information downloaded from this website.
Replace the battery if necessary. The Manufacturers reserve the right to fluuke this Information at any time without notice. Add to a parts list.
Rare Fluke 80T-IR Infrared Temperature Probe
You agree that the Information as provided here through an intermediary may not be error-free, accurate or up-to-date and that it is not advice. Accuracy figures are valid for a period of one year.
The product detailed below complies with the specifications published by RS Components. Please enter a message. Please select an existing parts list.
You have chosen to save the following item fluk a parts list:. Adjust the setpoint of the Blackbody to output each of the temperatures in Table 2 at 0.
Save this item to a new parts list. Verification Points Model Blackbody Temperature at 0. In between measurements 80r-ir the UUT to an environment not affected by the heat source. Record the reading displayed on the DMM.
Recommended Equipment Table 1 lists the equipment required to perform the calibration verification procedure. With the UUT powered on, check that the battery voltage is at least 7. Set the DMM to the 3. If your instrument fails to pass any performance specifications, return it to the Eluke Eactory or Service Center for repair.
The product does not contain any of the restricted substances in concentrations and applications banned by the Directive, and for components, the product is capable of being worked on at the higher temperatures required by lead—free soldering.